|Place of Origin:
|Minimum Order Quantity:
|3-20 working days
|T/T, Western Union, MoneyGram,Paypal
|S Type Force Sensor
Low Cost S Type Load Cell M12/M16/M18 Threaded S-Beam Force Sensor 0-50kN
Universal S-Beam Load Cell FS07 is made of alloy steel based on strain gauge technology intended for both tension and compression force measurement.Low cost S type force sensor FS07 offers 0-3kN / 0-5kN / 0-10kN / 0-20kN / 0-30kN / 0-50kN six rated measuring ranges to choose from,high accuracy with maximun 0.05% non-linearity of the full scale.M12/M16/M18 metric threaded holes located on both ends can be used to mount attachments such as eye bolts and hooks for different applications.S-type load cell FS07 offers an accurate and affordable measuring solution for applications like material testing machines, hanging scales, hopper weighing and so on. Model FS07 can also be modified or customized as per request of our clients.
300kg / 3kN / 660lb
500kg / 5kN / 1.1klb
1000kg / 10kN / 2.2klb
2000kg / 20kN / 4.4klb
3000kg / 30kN / 6.6klb
5000kg / 50kN / 11klb
|Temperature shift zero
|Temperature shift span
|Material of element
|Ø5.3*1800mm 4-core shielded
Load cell glossary:
Load Cell:A device which produces an output signal proportional to the applied weight or force.
Wheatstone Bridge:A Wheatstone bridge is a network of four resistive legs. One or more of these legs can be active sensing elements. The Wheatstone bridge is the electrical equivalent of two parallel voltage divider circuits. R1 and R2 compose one voltage divider circuit, and R4 and R3 compose the second voltage divider circuit. The output of a Wheatstone bridge is measured between the middle nodes of the two voltage dividers.
Excitation: The voltage applied to the input terminals of the load cell.
Rated Capacity: The maximum axial load the load cell is designed to measure within its specifications.
Non-linearity: The maximum deviation of the calibration curve from a straight line drawn between the no-load and rated outputs; expressed as a percentage of the rated output and measured on increasing load only.
Non-repeatability: The maximum difference between load cell output readings for repeated loadings under identical loading and environmental conditions.
Hysteresis: The maximum difference between load cell output readings for the same applied load; one reading obtained by increasing the load from zero and the other by decreasing the load from rated output; usually measured at half rated output and expressed in percent of rated output.
Creep: The change in load cell output occurring with specified period of time, while under load, and with all environmental conditions and other variables remaining constant; usually measured with rated load applied.
Safe Overload: The maximum load in percent of rated capacity which can be applied without producing a permanent shift in performance characteristics beyond those specified.
Ultimate Overload: The maximum load in percent of rated capacity which can be applied without producing a structural failure.
Operating Temperature: The extremes of temperature within which a load cell will operate without permanent adverse change to any of its performance characteristics.
Temperature Shift Span: The change in output due to a change in load cell temperature; usually expressed as a percentage of load per degree Celsius change in temperature.
Temperature Shift Zero: The change in zero balance due to a change in ambient temperature; usually expressed in percent of rated output per degree Celsius change in temperature.
Input Resistance: The resistance of the load cell bridge circuit measured at the positive and negative excitation terminals with no load applied and with the output terminals open-circuited.
Output Resistance: The resistance of the transducer bridge circuit measured at the positive and negative signal terminals with no load applied and with the input terminals open-circuited.
S Type Load Cell with M12/M16/M18 Threaded Hole S-Beam Force Sensor 0-50kN