|Place of Origin:||China|
|Minimum Order Quantity:||1pc|
|Packaging Details:||Shockproof carton|
|Delivery Time:||3-20 working days|
|Payment Terms:||T/T, Western Union, MoneyGram,Paypal|
|Type:||Press Force Laod Cell||Capacity:||0-20kN(Selectable)|
|Force Direction:||Pull And Push||Output:||Analog|
Flange to Flange Thrust Load Cell 0-20kN Press Force Transducer
Flange Mounted Thrust Force Sensor FFD is made of stainless steel based on strain gauge technology intended for both compression and tension force measurement, thrust force transducer FFD offers 0-2kN / 0-3kN / 0-5kN / 0-10kN / 0-20kN five rated measuring ranges to choose from, which delivers maximum 0.2% non-linearity of the capacity. Low profile design with only 25mm total height, flange-to-flange structure enables FFD to be easily and firmly connected to force applying mechanism,you may find press force load cell FFD suitable for applications such as hand press force verification, actuator force control, assembly force checking, spring testing, hydraulic cylinder thrust measurement, ultrasonic welding clamp force control,and so on.
2000N / 200kg / 440lb
3000N / 300kg / 660lb
20kN / 2000kg / 4.4klb
|Rated output||1.0 mV/V||Compensated temperature||-10...+40oC|
|Zero balance||±0.05mV/V||Temperature shift zero||±0.02% R.O./oC|
|Nonlinearity||±0.2% R.O.||Temperature shift span||±0.02% R.O./oC|
|Hysteresis||±0.2% R.O.||Input resistance||750±100 ohms|
|Nonrepeatability||±0.1% R.O.||Output resistance||702±5 ohms|
|Creep(10min)||±0.1% R.O.||Insulation resistance||>5000Mohms|
|Safe overload||150% F.S.||Ingress protection||IP65|
|Ultimate overload||200% F.S.||Material of element||Stainless steel|
|Cable||Ø5*3000mm 4-core shielded cable|
Red--Power input+ Black--Power input-
Green--Signal output+ White--Signal output-
A load cell (or loadcell) is a transducer which converts force into a measurable electrical output. Although there are many varieties of loadcells, strain gauge load cells are the most commonly used type which is comprised of a solid metal body on which strain gauges have been secured to form a wheatstone bridge which delivers mV/V output.
Load cell output: = (Power supply * Rated output) * Load / Capacity
Rated output of thrust load cell FFD is around 1.0mV/V.
Power supply is within 3-15V.
Load / Capacity is within 0-1.
So the output range of pull and push force sensor FFD is -15mV...+15mV(Tension and compression).
Example: Tension And Compression Load Cell FFD-20kN with 1.0mV/V rated output powered by 10V DC,the output range is -10mV...+10mV(Tension and compression).
A signal conditioner is normally used with the load cell to deliver -5~5V, 0-10V or 4-20mA output for easy data processing:
Example: Thrust Force Transducer FFD-10kN working amplifier FSC(0-10V)-24V,the load-voltage relationship will be 0-10kN thrust force VS 0-10V output,1V output corresponds to 1kN load, 2V output to 2kN load,and so on.
Flange to Flange Thrust Force Sensor 0-20kN Press Load Cell